Pro Check Thermometer
Weight: 1 thermometer
Important Safety: This device may be used only for measuring body temperature. The battery is well sealed inside the thermometer to reduce the risk to be swallowed by children. Protect the thermometer from impact and dropping. Avoid ambient temperatures above 140 degrees F. Never boil the thermometer. Use only the commercial disinfectants listed in the section Cleaning and Disinfecting to clean the thermometer. The thermometer must be intact when immersed in liquid disinfectant. Read the instructions carefully before using this device. Turning on the Thermometer: 1. To turn on the thermometer, press the on/off button. 1. A display test is performed. All segments should be displayed. Then at an ambient temperature of less than 90.0 degrees F, and L and a flashing degrees F appear at the display field. 2. The thermometer is now ready for use. Function Test: Correct functioning of the thermometer is tested automatically each time it is turned on. If a malfunction is detected, this is indicated by ERR on the display, and a measurement becomes impossible. In this case, the thermometer must be replaced. Using the Thermometer: Important: This thermometer can be used with or without a probe cover fitted. If a probe cover is used, fit a new probe cover on the measuring tip. If a Probe Cover is not used, cleaning and disinfecting should be done per the instructions. When taking a measurement, the current temperature is continuously displayed and the degrees F symbol flashes. If the degrees F is no longer flashing, this means that the measured increase in temperature is less than 0.1 degrees F in 16 seconds and that the thermometer can now be read. To prolong the battery life, turn off the thermometer by briefly pressing the On/Off button. Otherwise the thermometer will automatically turn off after about 8 minutes, 40 seconds. Measuring Methods: Oral Use: (for children four years and older). Do not drink hot or cold fluids, exercise, smoke, or perform other activities that will raise or lower temperature readings when compared to your normal, average temperature. Place the probe tip well under the tongue, (See diagram) as indicated by the check. The mouth should remain closed up to 5 minutes before attempting a reading. Instruct children not to bite down on the probe. The peak temperature should be reached in approximately 60 seconds. Opening the mouth or improper placement of the probe tip could result in a longer time for a reading. Underarm (Axillary Use): Wipe underarm with a dry towel. Place probe tip in armpit and bring arm down against the body. The measuring sensor should be pointing upward in the same direction as your body if possible. To receive more reliable results we recommend to measure temperature orally or rectally. Approximate measuring time for an axillary measurement is 5 to 10 minutes. Rectal Use: Commonly used for babies, young children or when it is difficult to take an oral or underarm temperature. After placing the probe cover on the thermometer, lubricate with a water-soluble solution such as lubricating jelly. Gently insert the probe tip no more than 1/2 inch or 2-3 cm into the rectum. Do not force the tip into the rectum if resistance is encountered. Dispose of the used probe cover and wash the thermometer as recommended. The peak temperature should be reached in approximately 60 seconds. Cleaning and Disinfecting: Use a soft slightly moist cloth clean the thermometer casing and the measuring tip. For disinfecting, wipe the thermometer with 70% isopropyl alcohol. Battery Exhaustion: When the symbol (upside-down triangle) appears on the right side of the display, the battery is exhausted and the thermometer needs to be replaced.
Model MT16K1. Technical Specifications: Type: Maximum thermometer. Measurement Range: 90.0 degrees F - 109.9 degrees F. Temp. Less than 90.0 Degrees F: Display L for low (too low). Temp. Greater than 109.9 Degrees F: Display H for high (too high). Measurement Accuracy: Plus or minus 0.2 degrees F. Operating Temperature: 50 degrees F to 104 degrees F; relative humidity of 15 to 95%; noncondensing. Storage Temperature: -13 degrees F to 140 degrees F. Battery: 1.5/1.55 V; SR 41. Reference to standards. Device Standard: Device corresponds to the requirements of the standard for electronic thermometer. ASTM E 11112; EN 12470-3; IEC 60601-1; IEC 60601-1-2. Electromagnetic Compatibility: Device fulfills the stipulations of the standard IEC 60601-1-2. Type BF applied part. Warranty: This thermometer includes 2 year limited guarantee after the date of purchase. Any damage caused by improper handling shall not be covered by the warranty. Batteries and packaging are also excluded. The manufacturer recommends a periodic calibration check every 2 years. Made in China.
This product is not a toy. Do not allow children to play with it. This digital thermometer is for home use only.
1617 Gulf to Bay Blvd 2nd Floor Clearwater, FL 33755 800-568-1417